SENSTREE G1/2" Liquid Display Water Flow Quantitative Controller Meter+Solenoid Valve
For DFC15 Controller:
Controller Power Requirement: 12VDC
Power Adapter: Input: 100-240VAC; Output: 12VDC, max 2A
Adapter lead length: 1m
Measuring Accuracy: ±1%
Output for Solenoid valve: 12VDC, max 5A
Temperature Sensor: 0-125℃/32-257℉, NTC3950, Accuracy: ±1℃/±1℉ thread: M8
Max Total Volume : 999999 G/L
Quantitative Range: 0-9999.99 G/L
Flow Rate Range: 0-9999.99/LPM(GPM)
Cable Length : 1m
Operation Environment: Temperature: 0-50℃/32-122℉; Relative Humidity: <85%
For G1/2" Flow Sensor:
Allow compression:Water pressure 1.75 Mpa below
Error: ± 2%
For G1/2" Solenoid Valve:
Operation mode:N/C,normally closed.
How to work:
1. Set K value, quantitative value, max alarm value, properly install the flow sensor and solenoid valve.
2. Press the RUN/STOP button, the controller works, total will continuously display the current flow volume, and current flow rate, and the faucet will start animated display.
3. When the flow reaches the quantitative setting value, the controller turn off the solenoid valve automatically, and the total value stop, faucet animation will be stopped dropping, flow rate back to zero.
4. Pressthe RUN/STOP buton again and the controller will repeat the same job.
1, This controller work with the solenoid valve up to 5A, but this kit come with 2A power adapter. If drive a large solenoid valve, please use suitable power adapter.
2, Ensure the solenoid valve and flow sensor direction is the same with water flow.
3, Avoid the water into the controller.
1 x DFC15 controller
1 x G1/2" flow sensor
1 x G1/2" solenoid valve
1 x Power adapter
1 x Temperature sensor
1 x Manual
We Also Recommend
G2" inch Water Liquid Controller Display Flow Control Meter Quantitative 10-200LPM
To date the order has not arrived.
this is the handiest thing i have ever had for my greenhouse... set and forget... no more spills. no more standing there watching water run!
Excelente servicio, el producto cumple con las características ofrecidas y la entrega fue antes de lo esperado
Once calibrated (using a weight measuring scale and adjusting the "K" Factor in settings) this meter appears to be rock solid. It come default "K" at 7.50. I have adjusted this to be more precise with a value of 6.92. Be interesting to know if others had to make the same adjustment Pay close attention to the connection harness as the strain relief on the cable is marginal at best. Mine was missing a screw and the other unit (I have 2 ) the wire jacket was not even clamped.
A cheap flow meter that can be connected to a solar panel and a battery leave it there working remotely for days. I used for a infiltration test in the middle of nowhere in central California. Keep it working in the field for a week in a very harsh environment and work great. I totally recommend this equipment.